Sonosite Engineering are presently working on a solution in order to support existing customers with SiteLink 7.
The Vet may not have ended the previous exam.
The scanner may be plugged into wrong usb port (Sitelink is port specific) - USB-Cisco adapter & training
You are not able to change the sector size.
In wrong application (abdo not cardiac)
Clean trackpad of all moisture using finger only, as a stray finger on the trackpad may confuse the scanner.
Enter setup mode by pressing the SETUP key. Click on the Audio, Battery menu and check sleep mode setting is off.
Backup images and clear CF card of all images.
Backup your images and clear the CF card of all images. Try another CF Card. Make sure that the CF card is in the front slot. The front slot is for images and the rear dock is for upgrades.
1. First, try using a different power lead.
2. Check fuse in power lead.
3. Order another mains supply.
1. Check battery age mm/yy.
2. Order new battery.
Check to make sure that you have the correct license key for the software and that you know the exact specifications of your software license.
Or login with emailForgot password